Install the device on a level surface.1
As a general rule, the device is installed on a level work bench in the measurement area.
Prepare the standard block.2
Pull the standard block from the organizer box.
Standard block (25mm)3
This is the standard block (25mm). There may be dirt or something may be sticking to both edges of the standard block, so wipe it with a clean cloth before use.
Measure the standard block.4
Position and measure the standard block. To tighten, twist the tip of the micro meter as shown. (If you twist the thick base, it may be overly tightened.)
Slightly tighten it.5
A zero point adjustment.6
In the case that it does not show 25mm when tightened, need a zero point adjustment.
Press this button twice.7
Make a zero point adjustment, press the button labeled
Measure the target object of measurement.8
Measure the target object of measurement with
the micro meter after making the zero point adjustment.
Make sure to measure the largest part of the outer9
When measuring the outer diameter of circular forms object, make sure the position of the micro meter is on the largest part of the outer diameter.
Adjustment for the correct result10
Depending on the shape of the object,
you may have to adjust the angle and position
of the object to get the correct result.